Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||15 October 2009|
|PDF File Size:||10.19 Mb|
|ePub File Size:||11.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set the spectrum analyzer as foflows: Press the Tuning Const key to select Compute from expected T. Set the carrier frequency of the Signal Generator to O iV at MHz 6.
Full text of “HP: B B final”
When the Signal Generator is turned back ON, the instrument functions laist displayed will be active, even if the plug has been disconnected. Provides information to allow you to use the feature or function. The Signal Generator automati- cally performs a self-test check at power on this takes approximately 6 seconds. Pulse Modidation is available as an option on the HP B only. These specifications are the performance standards or limits against which the instrument may be tested.
Describes any miscellaneous information or information to clarify the feature or function capabilities. This sets the two input signals to the mixer in phase quadrature. Slide rack mount kits are discussed in the following paragraphs; refer to paragraphMechanical Options, in Section 1 for information and part numbers pertaining to other rack mount kits. If the Signal Generator is in local operation when it is enabled to remote operation that is, REN is manuual true and it receives the Local Lockout message, ph switches to remote with local lockout operation the first time it is addressed to listen.
Table lists the power consumpdon of each Signal Generator. Frequency HP B If external reference is used, ouput mil the same frequency. The Signal Generator must have a minute warmup.
Verify that the measured frequency does not exceed the Umits specified.
A 3-digit display of AM depth or FM peak deviation is provided with 12 LED annunciators used to display internal or external modulation source information.
Comments You can save up to complete front-panel setups exclusive of increment settings. Connect the digital multimeter to A13J2 pin4. HP B HP Connect the digital multimeter to the test points shown in the table below and adjust for a 0. Switch off the HP B and remove the power cable. Comments The currently selected modulation function AM or FM is turned off when the OFF key is pressed, provided only one source internal Hz or 1 kHz, or external is selected for use.
The Service Manual contains information to enable qualified service personnel to troubleshoot and repair the Signal Generator.
It calls attention to an operating procedure, practice, or the like, which, if not correctly performed or adhered to, could result in damage to or destruction of part or all of the product.
Verify that the Line Voltage Selection Cam is matched to the power source. Frequency Span settings on the ,anual analyzer can be reduced to as low as Hz and Center Frequencies changed to facilitate viewing the spurious signal. The following steps mil guide you through the protxss of making the necessary changes to the parameters to meet the specific requirements of this test.
The maanual switch segments are used to invoke built-in service utilities. Constant Press the Done key. Hewlett-Packard offers a Documentation Update Service that will provide you with further updates as they become available. Rotate cam so 8656bb the desired tine voltage appears at the Cover Door Window when the cam is re-inserted.
The slide kits for the Signal generator are listed below. Ideally, the frequency change should be made from the same preset frequency. NOTE This maunal only verifies level fiatness, it does not verify absolute level accuracy.
Ensure that the Signal Generator has the correct line voltage selected and that the correct fuse has been installed. Verily that all harmonics are within the specifications shown in the following table.
HP 8656B Manuals
Set the output amplitude of the Signal Generator to 0,0 dBm, carrier frequency to 0. Detailed Operating Instructions Dlnstrations in this section of the manual, provide you.
Performing this feature does not affect the current set-up. Switch off the Signal Generator and remove the power cable. Set the output amplitude of the Signal Generator to — AM Hates Internal and 1 kHz. Tune the frequency of the spectrum analyzer for maximum level on the display. The two MHz Notch Hlters are in the circuit all of the time. If the Signal Generator is in local lockout when the Local message is received, full front-panel control is returned, but local lockout remains in effect.
Increase the output of the test oscillator by rotating the Amplitude Coarse control clockwise until the measuring receiver indicates HP HP Procedure: Refer maunal Section 2 2.